Pupillary axis, kappa angle
The pupillary axis is year axis of importance, as it serves to define several angles. The pupillary axis is defined as the line that joined the center of the entrance pupil to a point located at the anterior corneal surface, where the local tangent is perpendicular to it If the corneal was perfect spherical dome aligned with the physical iris opening (iris pupil), there would not be a 'unique' pupillary axis.
The corneal surface is irregular and the pupil is not aligned with the cornea. Hence, there is one pupillary axis which can be found by asking the patient to fixate monocularly (with the eye of interest) a non moving target, and display a bright point source of light until its reflection appears for year observe coaxial with the source located in the heart of the entrance pupil.
The pupil axis serves to define the kappa and lambda angles. The kappa angle is defined as the angle between the pupil axis, and the visual axis; In some treaties, this angle is called angle "Lambda" (if so, angle Kappa labels the angle between the pupil axis and the line of sight). In practice, the values of kappa and lambda angles are close: remember that the distance between the respective corneal intercept of the line of sight and the visual axis is (despite clinical relevance) short (a few hundredth of a millimeter compared to the distance between these points and the corneal intercept of the pupil center.)
As this intercept rental is not measured by clinical routine testing, The Orbscan instrument may be the only topographic device to provide year estimate of the true kapa angle, as this instrument is capable of measuring the distance from the cornea to the entrance pupil and locate its center.
Pupillary axis serves as a reference axis for the kappa angle:
The pupillary axis is not necessarily located in the plane formed by the line of sight and normal setting (line connecting the fixation target and the first Purkinje image i.e. the reflection of the target by the corneal surface, not shown in the above diagram).
Other pure specular Placido based topographers can only measure the curvature of the front surface of the cornea: from the distance between images of the vertex and of the center of the entrance pupil. This distance is sometimes improperly refered as the "kappa corner". The reason for this (incorrect) assumption comes from the fact that the larger the Kappa angle, the larger the distance between the specular reflex of the coaxial fixing source and the center of the entrance pupil.
This year is example of the influence of the Kappa angle on the first Purnkinje image location (the camera of the IOLMaster 700 is coaxial with the center of the fixation mire which is used to calculate the keratometric power).